Company News About Every foreign trade person should know about liquid level sensor products
Every foreign trade person should be aware of the matters related to liquid level sensor products, and if it is not divided into standard and non-standard products, everyone can choose through standards or standards. However, even though iPlayer is committed to producing various types of liquid level sensor, many people are still unaware of some misunderstandings about liquid level sensor.
In fact, the misunderstanding of the split type liquid level sensor is that it cannot be damaged by the liquid level sensor. Many customers often confuse incorrect split type liquid level sensor due to misunderstandings about sensor.
Before the application of the split type liquid level sensor, unlike the split type liquid level sensor, the wiring terminals and cables of the split type liquid level sensor were used for operational amplifiers. Therefore, it is necessary to inquire about and correct the parameters and specifications of the split type liquid level sensor in order to make a correct misunderstanding of the split type liquid level sensor.
The cross issue is that when most split level sensor are damaged, the zero and related values may not be appropriate. Some data will deviate from the terminals and cables of the split type liquid level sensor, changing the zero position value of the split type liquid level sensor. The cross issue is that when most split type liquid level sensor are damaged, the zero value will exhibit the following abnormal bending phenomenon.
Split type liquid level sensor and split type liquid level sensor have different forms of liquid level split type liquid level sensor in different sizes.
Firstly, if there are no gaps on the surface of the gauge eye, but the liquid level cannot be measured, a magnetic flap level gauge should be used to measure the reading. Observe the strength of the surface eye. If there are many situations where the eyes are dry and there is a bit of water fragmentation, even some liquid level detectors may not have the following effects
The magnetic flap level gauge cannot display the liquid level and does not work at high speed. Sometimes, manual errors or data errors may occur, and it usually needs to be replaced.
Secondly, use a multimeter to measure its range. If there is a digital display that cannot display as, it is likely to result in incorrect readings. If it is a mechanical button, it is very likely that the internal resistance of the multimeter will not be blocked. It is necessary to unscrew the wiring inside the junction box of the multimeter and remove this board.
The magnetic flap level gauge frequently gets stuck in the upper and lower water pipes, resulting in jamming or damage. Regularly inspect and replace damaged detectors or replace them with new ones.
Some liquid level detectors (such as float switches) have reverse protection coefficients and are not set correctly. Solution: Remove the junction box and reconnect it to fix it, leaving the float switch in a power off state.
How can we choose the correct installation method for float switches? We should take the installation proposal seriously and have a good understanding.
When installing, first place the float switch from the measuring tube and then remove it from the instrument. Be careful not to move the instrument float.
When the measuring medium is a fluid that is prone to crystallization or sediment, sealing gaskets should be added to the pipeline and attention should be paid to preventing backflow. The pressure switch should not be affected by water vapor and atmosphere.